Statistics from Altmetric.com
We introduce a novel, proof-of-concept polyp measurement device for the design of future colonoscopes. In a simulation study, 50 expert endoscopists used three measurement strategies (method 1: no visual cues; method 2: a snare tip visual cue; method 3: a grid visual cue) to evaluate 120 images of 40 simulated polyps (1–10 mm) presented in random order. Measurement accuracies of each method were 20%, 33% and 90%, respectively (p<0.001). Using the proof-of-concept measurement grid (method 3), endoscopists were 99.8% accurate for high confidence classifications of polyps into clinically relevant size categories. Improvements in colonoscope design are needed to minimise human and technology biases.
In more detail
There are increasing concerns about the influence of human bias on the measurement of polyp size at colonoscopy.1–3 A study of 92 124 individual colorectal polyp measurements from the UK bowel cancer screening programme showed that endoscopists, radiologists and pathologists all exhibited terminal digit preference and clustering of polyp measurements at 5mm intervals.1 Although human bias is an acknowledged contributor to the measurement error of polyps, the importance of technology bias remains understated.1–3 Figure 1 illustrates the extent of image distortion from the fish eye lens of modern colonoscopes, where objects in the centre of the display appear magnified and objects at the periphery appear small and warped. It also demonstrates how makeshift measurement tools, such as snare catheters, potentially worsen measurement error from technology bias coupled with illusions of relative size. We hypothesised that both human and technology biases could be addressed by improvements in current colonoscope technology.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.